- SONY ICD-B16 User manual - Nodevice
- SONY ICD-B26 OPERATING INSTRUCTIONS
- Sony eSupport - Voice Recorders - Support
- SONY ICD-B16 SPECIFICATION SHEET Pdf
SONY ICD-B16 User manual - Nodevice
If the battery compartment lidis accidentally detached,attach it as illustrated. However, there is no guarantee that interference will not occurin a particular installation.
SONY ICD-B26 OPERATING INSTRUCTIONS
Record the serial number in the space provided below. Step 1: Installing the Batteries Slide and lift the batterycompartment lid. HH: To record at meeting or in a quiet and/orspacious place. * About one hour after you pause recording, recording pause is automatiy released and the unit goes into the stop mode. B Basic Operations Remaining memoryindicator * The display selected with the DISPLAYbutton appears. These limits are desned to provide reasonableprotection against harmful interference in a residentialinstallation.
Sony eSupport - Voice Recorders - Support
Also some problems and solutions are described in this document.
SONY ICD-B16 SPECIFICATION SHEET Pdf
3-264-832- 11 (1) IC Recorder Operating Instructions 2004 Sony Corporation Printed in China ICD-B16/B26 Owner’s Record The model number is located at the rear and the serial number is located inside the batterycompartment. Note Before making a long recording, be sure to insert new batteries and check the batteryindicator. recording instantly select the microphone You can select the microphone sensitivity to display sensitivity the following by pressing MIC SENS. The unit stops at thebeginning of the currentrecording. INFORMATION This equipment has been tested and found to comply with thelimits for a Class B dital device, pursuant to Part 15 of the FCC Rules.
Manual grabadora sony icd-b16:
Rating: 97 / 100
Overall: 93 Rates